This calculation model is only valid for SRD
Model Name Brix: Valid for SRD
1. What it does?
Determines the pure sucrose content in water from line density and temperature
Soft drinks, fruit juices, tomato concentrates – Beverage Industry - Cutting Oil – Machining/Industrial.
Below finds the mathematical formula for Brix degree calculation.
To obtain the correct measurements for Brix degree, the SRD and SMET work together as a powerful tool for measuring density, which is an extremely significant variable. The user would need to input:
- Values dataset of densities of different products at different temperatures
- Input Water density at T2 (20°C – brix). Needed for Specific Gravity @20°C calculation. Compensated density is calculated with the values measured from the SMET.
- Measure line density and line temperatures. Values are taken from SMET-SRD.
- Convert the line density of process fluid to temperature compensated for density value at T1 (20°C –brix) – By using a dataset table from known products.
4. How to load models into the SMET?
The calculation tab allows configuring the SMET to run mathematical models for Viscosity, Density, and Concentration.
Figure 1. Calculation tab for Brix degree model
To load the models into SMET, follow these steps:
5. Matrix Referral
Figure 2. Sample Dataset for Brix readings.
- Dataset for beverage, the first column is for temperature. The rest of the columns should be each for a product "P", and the densities of that product at the different temperatures.
- Import: Select dataset from an .csv sheet
Figure 3. Selecting dataset
3. Export: Export the current table to .csv
4. "Tk1, Tk2" button to create the coefficients that will be used on the model.
5. Calculated Tk coefficients by internal model.
6." Load Tks & Close" button to load these coefficients into the model
6. Uploading the Mathematical Model
Click on "Upload Model" to load the model settings into SMET. If the operation was successful, the button should turn GREEN for a couple of seconds; otherwise, it should turn RED. This action will refresh the values for the Uncompensated and Compensated indicators. Below is a sample of Brix at the beginning of a fermentation process along with a typical fermentation curve demonstrating the full capabilities of the SRD monitoring system.
Figure 4. Brix model loaded into the SMET.
6.1. Click “Load Configuration” to update the display and channel configurations in SMET.
- LCD Display
Select the line to display the parameter:
Figure 5. Selecting read line for LCD display
Changes can be verified on the “Communication” Tab
- Analog outputs
Select analog channel to send data, this can be verified in “Service” Tab
Figure 6. Selecting analog output channel
- Modbus TCP/IP register
Measurement can always be accessed at the register address that is displayed.
- “Calculation History” button allows you to open a file where the last uploaded models and coefficients for these models are stored. This is for user reference.