Which Rheonics products are related to this Article? This Article is based on the use of the SME (Smart Module Electronics) from SRV, SRD, DVP and DVM. Hence it can be used for any of these Rheonics sensors.
What is the purpose of this Article? To give guidance and good practices when doing electrical installations and wiring for Rheonics Sensors and SME.
Typical Wiring Scheme
Figure 1: Typical SME-DRM Installation.
1. General Considerations
1.1 Minimum distances for power lines and control cables
Keep a minimun distance of 50mm (2”) between power and control/signal cables.
To avoid pulling tension problems and keep strain relief:
Attach the pulling line in the correct manner.
Avoid exceeding the maximum tensile strength of the cable.
Relieve the pulling tension once the cables are in place.
Figure 2: Pulling tension on cables.
1.3 Bend Radius
Consider a bend radius for cables. Fixed installations require a minimum bending radius (R) of five times the diameter (D). A minimum of ten times the diameter (D) is recommended for flexing/moving installations.
Figure 3: Bend Radius for Cables.
1.4. Cables Crush Resistance
To not exceed the cables crush resistance consider:
Figure 4: Hook and Loop Fasteners for Cables 
- Do not step on or run over cables with vehicles or equipment.
- Be aware of other areas that may constrict cables or conductors and put them at risk of damage.
- Communications cables should not be stapled with standard staples, which may crush them. Use cable staples of the correct size.
1.5 Temperatures Rating for Cables
Specially for installations on harsh enviroments, with high temperatures, cables in possible contact with fluids, or outdoors applications, consider a cable sheath material capable to handle the temperature rating.
Standard sensor cables supplied by Rheonics have a temperature rating of -30°C to 90°C. For different requirements, user needs to notify it in the RFQ (Request for Quotation).
1.6 Cables Gauge connected to the SME
Figure 5: Crimper for Wires.
Cable Sheath Properties
The cable’s sheath should be selected based on the installation place, liquids cables are exposed to, etc. Next Table show principal sheath materials for industrial applications:
2. Cables involved in a typical Rheonics SME Installation
The SME-DRM has the following terminal blocks involved in a typical installation:
Figure 6: SME-DRM Terminal Blocks.
2.1 Power Cable:
Positive and negative terminals are connected to the SME through push-in terminal block 1.
DC or AC Power Cables can be single or multi-core cables, depending of the need of flexibility, space available and amount of devices to supply voltage.
Power cables, AC voltage specially, must have a considerable distance from signal, control and communication cables, around 50mm minimum. When using wire ducts inside an electric panel, power and signal cables are recommended to go through different paths, even if they go to the same device.
2.2 Analog Signal Cables:
The SME has 3 non-isolated 4-20mA output channels, placed in terminal block 2.
Selecting a good cable here is important to ensure a "clean" signal to make its way from the SME to the PLC or analog reader.
Always use stranded wires (ideally with ferulles or crimpers on the tip). These wires offer good flexibility and bendability, making the installation easier and cleaner.
Correct Gauge for the SME's Analog Signal can vary between 18, 20 and 22AWG. Consider that 4-20mA signals are affected by a harsh enviroment and the lenght of the cable. The bigger the gauge, the less signal loss and resistance ocurrs, but the harder the cables are to handle.
To face the harsh enviroments of industries, shielded cables are neded for these signals. This shield is a metallic armour that prevents electromagnetic fields, created by the flow through the wires, to come out of the cable and avoids other fields to reach the wires inside the cable. Depending on the level of noise, the shielding options (ordered from low to higher protection) can vary from: Foil Shield, Braid Shield and both at the same time.
Twisted wires (wires wrapped around each other) are also a useful characteristic that creates electromagnetic compatibility (EMC) between wires of the same circuit and can reduce magnetic noise coming from mechanic or electric devices.
Figure 7: Shielded Cables 
2.3 Ethernet Cable:
Ethernet cables with RJ45 connectors can be used for multiple communication protocols with the SME and another device (e.g. PLC, PC, etc.). When using it, consider a minimum bending radius of 25mm. Depending on the length of the connector itself, the minimum distance “n” in the next drawing may vary but should be close to 70mm.
Figure 8: Ethernet Cable
SME’s ethernet connection should go to a ethernet switch where other devices can be connected to create a network.
2.4 Sensor Cable:
Sensor cable is included when a Rheonics sensor is purchased. It has, in one end, a M12 8-Pin-Female connector that should be inserted into the M12 8-Pin-Male connector of the Sensor. While in the other end, it has 8 stranded 20AWG wires crimped with white ferrules, that should be placed in the Block Terminal 3 of the SME in a specific order by the colors, see article:
Sheath’s material, cable length and connection orientation (0° or 90°) are customizable when doing a purchase.
Figure 9: M12 0° Connector 
Figure 10: M12 90° Connector 
2.5 USB Cable:
DISCLAIMER Pictures, photos and images used are for illustration purposes and do not constitute any warranty or assertions on suitability of use and should not be construed as an explicit or implicit recommendation or endorsement. All illustrations are given proper credit from our source of access and by their use here we do not specify or establish any copyright which belongs and stays unaffected with the existing copyright holder of that material.